Date/Time Issued: January 09, 2019 0102Z
Product(s) or Data Impacted:
All Products from visible Channel
Date/Time of Initial Impact: January 08, 2019 0930Z
Date/Time of Expected End: Until Next Month's Coefficient Update
Length of Event: N/A
On 08 January 2019, the following update of the calibration coefficients, based on post-launch vicarious calibration at NOAA/NESDIS/STAR, was implemented for the Imager visible channel of GOES-15 at 0930Z. These coefficients are intended to be optimal on 22 January, two weeks after the implementation date.
This notification no longer includes updates to the Imager visible channel calibration coefficients for GOES-13, since GOES-13 is no longer GOES-EAST.
The following update of the calibration coefficients, based on post-launch vicarious calibration at NOAA/NESDIS/STAR, is recommended for the currently operational GOES-15 Imager visible channel. These coefficients are expected to be implemented on the second Tuesday of the Update Month listed below and optimal on the fourth Tuesday of the Update Month (in the middle of the two scheduled updates).
Update Month: January
Update Year: 2019
Current post-launch calibration coefficient: 1.57923
Previous post-launch calibration coefficient: 1.57470
Initial calibration coefficient: 1.0
Initial calibration date: 2001114000000000
The date format is: YYYYDDDHHMMSSmmm, where YYYY = year, DDD = day of the year, HH = hour, MM = minute, SS = seconds, mmm = msec.
The coefficient provided is a gain that is to be applied to the Visible channel counts-to-radiance conversion
Op_R = Coefficient * Pre_R
where Pre_R is pre-launch gain for the channel and Op_R is the new gain -- the gain after the coefficient listed as "Current post-launch calibration coefficient" is applied.
The initial coefficient has been normalized to 1.0 for the purposes of this post-launch calibration.
Contact Person(s) Name/Email/Phone Number for
Paul Douglas, email@example.com, desk phone 301-817-4031operations support
Xiangqian Wu Xiangqian.Wu@noaa.gov <mailto:Xiangqian.Wu@noaa.gov> 301-763-8136 x 138 instrument scientist
Web site(s) that contain other relevant information (where applicable):http://www.star.nesdis.noaa.gov/smcd/spb/fwu/homepage/GOES_Imager_Vis_OpCal.php
Contact Information for Further Information:
ESPC Operations at ESPCOperations@noaa.gov
This message was sent by ESPC.Notification@noaa.gov.
You have been sent this and other notifications because you have
opted in to receive it. If for any reason, you wish to
unsubscribe, please contact ESPC Help Desk at ESPCOperations@noaa.gov
or (301) 817-3880. Please note: it may take up to two business
days to process your unsubscribe request.